00 BEDS HEALTHCARE FACILITY INCENERATOR TECHNICAL INFORMATIONS
– – Waste disposal administration will be taken unique note to prevent health center infection.
– – Infections wastes shall be taken care of, stored purely and also burned by a burner situated at hospital site.
2) Waste Disposal Plan
– – Medical wastes in healthcare facility are categorized as follows;
|Infections Waste||Specially-controlled waste||Organ, needle, etc.|
|Industrial waste||Rubber gloves, etc.|
|General waste||Gauze, etc.|
|General Waste||Plastic, Rubber||Infusion bottle, etc.|
|Paper, wood, Vinyl||Voucher, etc.|
|Metal, Glass||Can, Glass bottle|
- b) Infections Waste Disposal
– – Partition: Infections garbage disposal will be set apart suitably and also crammed in particular containers as well as delivered to particular storage space. The partition, packaging and distribution will be run appropriately at each room such as outpatient, operation cinema, blood collection area and also dialysis space.
– – Safety measure: Infections waste will not be location in other containers on the way of partition, packing as well as delivery.
– – Storage space: Infections waste will be kept at the storage space in health center website briefly until infections waste is incinerated.
– – Incinerator:
It is developed to sanitize clinical waste, decrease its quantity, and make its parts unrecognizable all in one completely enclosed and automated system.
It combines straight heated steam, high stress and also shredding to deal with contagious materials.
The infected waste is filled right into the top stainless-steel chamber where a sturdy shredder reduces the waste into tiny items. Utilizing gravity, the processed material comes under the therapy chamber where every item is heavy steam warmed to 134 and also pressurized to 3 to bar for 20 mins. The operating conditions are continuously kept track of as well as validated to achieve complete sterilization.
The 60 mins typical duration procedure cycle is totally automated. A computerized system monitors the procedure and also instantly prints a record with the crucial disinfecting criteria at the end of the cycle for exact record keeping.
The ability is between 500 to 600 kg/cy. h or even more.
1- Filling; the polluted waste is automatically -filled into the chamber of the system
2-Shredding; the durable shredder reduced the product right into small pieces and also includes a special turning around system to avoid jamming
3-Heat; saturated pressurized heavy steam increases the temperature to 134 C and the pressure to 3 bar
4-Sterilization; the sterilization enters straight call with the shredding material (134 c, 3 bar) for 20 mins
5-Cooling; temperature level is decreased to 80 C by spraying awesome water onto Double coat of the treatment vessel, concurrently, the unit returns to ambient pressure
6-Draining; the heavy steam is condensed to water and released with the cooling water to the sanitary sewer system
7-vacuum; the staying residual steam is vented out through an air pump
8-Unloading; a digital signal educates the operator that all safe operating conditions have been fulfilled
The problem unlocks the device and also permits gravity dumping of the cured waste right into a container
|Operational temp. max||160 C|
|Max Weight when filld with waste For a special hydraulic test. Kg||10500|
|Stress. Kg/ cm3||3.5|
|Steam Pressure. Bar||8|
|Max Steam Flow Kg/h||500|
|Compressed Air Bar||6|
|Electricity 380 V/3 – phase Surge Protection system||35KW|
|Average Cycle Time. minutes||60|
|Process Wight Capacity . Kg||500 – 600|
|Average process wight capacity Kg./cycle microbial inactivation||200-300
|Waste volume reductin||80%|
|Consumption / Cycle|
|Electricity . KW/h||9|